Welcome to our site. We have tried to make it as comprehensive
and useful as possible. Monitor’s discipline, the backbone
of our company ethic, is “Practical solutions...at every level!”.
Our goal is to always deliver to our customers a combination of product
quality, price and ease of purchase that no one else in our industry
can match. Maintaining
customer loyalty, and generating good “word of mouth” is
vitally important to Monitor’s day to day business. Here in
the "Showcase" area, we have provided links to new
product news, success stories,
seminars, white papers, and a what's
to get up to date with Monitor's happenings. If that isn't enough,
check out our Trade
Show schedule and meet one of our representatives in person.
Please bookmark our site and visit us often. New
information is constantly being added.
Monitor Technologies, we have the solution for you!
Advanced RF Capacitance Level Probe
The TrueCap® Model MK-2 RF Capacitance
point level probe is designed to provide a superior and stable sensitivity
threshold making it suitable for a variety of powder / bulk solids
and some liquid or slurry applications.
Advanced features of the
Model MK-2 include: > Automatic immunity to material build-up on the
probe by its driven shield design > Push-button calibration > Enhanced temperature
Maximized reliability via smart sensing algorithms like “self-validating” fail-safe
protection > Visible status LED on ordinary location
units > Versatility through a variety of configuration
options including: hazardous location version, split architecture
design, quick-connect process connection, stub probe, cable extensions,
solid extensions, Nylon® probes, Ryton® - equiv. probes,
A practical application for the TrueCap would be to use this level sensor where
a residual material build-up on a different sensor would cause a false
material level indication.
Principle of Operation for the TrueCap RF Capacitance Level Probe:
wall and the active probe element establish an impedance reference between each
other when exposed to air which has a dielectric constant of 1. When materials
with a dielectric constant greater than 1 are in close proximity to the probe,
the impedance of the sensing field between the sensor and the vessel
wall will change. Once the amount of change exceeds a threshold that
was electronically determined during the calibration process, an
output relay will either be energized or de-energized depending upon
the position of the fail-safe selector on the probe’s electronic
circuit board. A change of as little as .5 pico-farad is all that
is necessary for the probe to sense the presence of material.